Every time the Tricolops Dimensioning application starts up, the trigger area needs to be re-calibrated.
The most common causes are:
The scale has moved outside the original trigger area.
There are three options:
Move the scale back into its original position and the dimensioner will automatically re-calibrate.
Decrease size of trigger area so the scale will not move outside of the trigger area.
Go to Advanced -> Options, scroll down to Advanced tab, and uncheck "Auto Calibrate".
The dimensioner has moved. Use the provided hex key and tighten all connecting clamps.
Add a Comment
About This Community
This forum is actively monitored by our support and engineering staff. Read Guidelines
|Asked: 11/26/18, 4:32 PM|
|Seen: 1456 times|
|Last updated: 1/28/20, 4:25 PM|